Dec 31, Operating and service manual for the HPA Synthesizer/Function Generator. Complete with schematics. Who I HEWLETT mllnM PACKARD OPERATING AND SERVICE MANUAL MODEL A SYNTHESIZER/FUNCTION GENERATOR Serial Numbers: AH. HP / Hewlett Packard / Agilent Used A Synthesizer/Function Generator, Refurbished – Used Test Equipment at Test Equipment Depot.
|Published (Last):||13 January 2008|
|PDF File Size:||20.38 Mb|
|ePub File Size:||17.13 Mb|
|Price:||Free* [*Free Regsitration Required]|
Triangle and Ramp Verification. This section contains the procedures required to adjust the A to meet its specifications in 33255a Equipment required to maintain the Model A is listed in Table Devices which communicate along the interface bus fall into three basic categories.
HP 3325A Calibration Manual
Without changing the reference level, change the spectrum analyzer center frequency to 5. Remote control of the A by means of an HP-IB system controller is accomplished through this connector.
Record the voltmeter reading in the 3 V sine wave 1 kHz reference space on the performance test record. Amplitude Modulation Distortion a.
Power consumption is VA maximum. Set the dc offset to the positive and negative voltages shown below. Set DC voltmeter to IV range and ext trigger.
HP A User Manual | 23 pages | Also for: B
Configuring, Polling, and Unconfiguring. Fully Programmable All necessary functions are programmable on the HP-IB, including frequency, amplitude, all functions, phase and dc offset, modulation, all sweep parameters, amplitude cal and self-test, making the HP A a very versatile and powerful addition to automatic test systems.
Set A frequency to 2 kHz and amplitude to 40 Vpp. Maximum and minimum amplitudes are shown in Table In any correspondence, refer to the instrument by model number and full serial number. Press the high voltage key near the A output connector. Change A amplitude to 50 mV p-p, and change oscilloscope vertical to.
High Voltage Output Option Connect an ac digital voltmeter to the A signal output. After receiving an interrogation, the A will send back the following the next time it is addressed to talk: The system controller may designate itself as either talker or listener.
Full text of “HP A Synthesizer/Function Generator Operating and Service Manual”
Harmonic Distortion relative to fundamental Fundamental Frequency 0. Change oscilloscope horizontal control to. This is the flatness reference voltage. Items of equipment used in this procedure are referenced within the text by common name as listed in table 2. Set CRO as follows: Adjust Offset In A3R33 for a voltmeter reading of 0.
Providing the finest test equipment solutions since 1992
Neither light on, indicates the front panel signal output is active. If the program string contains 48 characters or more, the A will hold up the bus while it processes the 48 characters before accepting and storing the 33325a of the string. Power Line Voltage Selection Procedures for checking electrical performance are given in Section IV.
Electronic counter should indicate 20 This key must be pressed to select any of the key functions labeled in blue. Sine Wave Amplitude Calibration. This procedure checks the voltage levels of the sync output square wave: The HP A provides unprecedented performance per dollar thanks to 33225a major contributions from advances in HP technology. After interrogation, the error code is set to zero until the next error occurs.
Meta Messages Block Diagrammed. Bump the Manaul frequency to